Before you When calculating rates.g. Quick Start-Up Guide Setting up an infusion for the Ivac Rate Calculation ml to be infused in 2 hours 30 minutes. The IVAC® Volumetric Pump automatically regulates the infusion rate of intravenous solutions. Users are advised to read, to understand this manual and to be thoroughly familiar with the Pump prior to Tel: (27) (0) interferon α-2α, , , , , , , interferon α-2b, , , , , IVAC, ixabepilone, –,, ,, , ixazomib, .
|Published (Last):||18 January 2016|
|PDF File Size:||16.51 Mb|
|ePub File Size:||14.97 Mb|
|Price:||Free* [*Free Regsitration Required]|
Remove the mamual sealing cord from the groove in the lower case. See Chapter 3, ‘Routine Maintenance’ for details. Allow the pump to run until it touches the probe and then using a stop watch, time the travel over a distance of Disconnect the clip securing the flexi circuit to the base of the chassis plate see Figure 3.
Front panel labels cannot be reused. Maximum value cannot be set below the minimum value. If you wish mannual discard electrical and electronic equipment, please contact your CareFusion affiliate office or distributor for further information. Check maual the pump is travelling at the correct speed, within the range of 4mins 50secs to 4mins 58secs.
If a torque driver is available for servicing this will help control the applied torque; otherwise, be aware that excess force may cause the component to fail. Enter a File name i.
CareFusion IVAC 598 Directions For Use Manual
The internal rechargeable sealed acid lead battery will retain charge if maintained correctly. Keyswitch The keyswitch has three positions. Turn the pump ON. The position at which the display changes is the EOI point iavc it depends on the type of syringe in use. The washer and the rod support will still be on the rod. R Remove the screw and washer securing the potentiometer assembly then remove the potentiometer assembly.
S tep through each DRUG option see table below and modify as required. Please contact a local customer services representative kvac obtain login details. See ‘Entering an Access Code’ for instructions.
Lots in Sale Number | Hilditch Group
Retain the fixings for reassembly. Table of Contents Add to my manuals Add. Siemens SC Ventilator with Bracket. Hinge the base plate open and remove it. Ensure there is no excess lead length from the capacitor and that the capacitor leads do not mankal any adjacent components. A serial to parallel adapter is required. Interference may occur in the vicinity of equipment marked with the following symbol: Remove the syringe shaft assembly from the upper case. Continuous infusions are not available.
R emove the three pole clamp screws and washers. These notes provide useful advice or information that may help to perform the task more effectively. The indicator light can be configured to show when the PCA dose is available or being delivered.
Power Supply Supervisor The power supply voltage is monitored by IC6 which provides an active low reset signal to the microcontroller at power up and if the regulated 5V input falls below a preset level. In parallel to the patient controlled operation the pump allows the clinician to set an automatic loading dose which will be delivered at the onset of treatment.
The green LED will flash during infusion. Search this sale Search all sales. Fit the linear test gear TG and JG plunger protect to the pump, and TG magnet to the cover clip position and move plunger holder towards the dial gauge until it is a about 3mm clear of the probe. Plunger Detect The syringe plunger button position is detected using a slotted optical switch.
This signal is applied to a schmitt trigger, IC14, via PL5 and the output is then read by the microcontroller via IC9. The pump does employ an internal short duration battery. Remove the handset assembly. Key Switch Key switch: Valleylab Ligasure Vessel Sealing System. Fuse FS3 provides a separate ground connection for the audible alarm such that if fuse FS2 were to blow when the pump is running on the internal battery, a continuous alarm would sound.
Remove the securing pin, whilst supporting the motor shaft. This key lock is located on the left side of the pump SM Iss. Wherever this symbol is shown an Update note is found. Remove the three retaining screws and washers. If the pump is dropped, subjected to excessive moisture, humidity or high temperature, or otherwise suspected to have been damaged, remove it from service for inspection by qualified service personnel.
In Volume mode – between 0. Fit replacement cover where required.