View and Download HP A calibration manual online. Synthesizer/Function Synthesizer HP A Operating And Service Manual. Frequency ( pages) . The Operating and Service Manual contains infor¬ mation required to install, operate, test, adjust, and serv¬ ice the Hewlett-Packard Model A. Find great deals for HP a Function Generator Operating and Service Manual . Shop with confidence on eBay!.
|Published (Last):||1 July 2013|
|PDF File Size:||9.74 Mb|
|ePub File Size:||19.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
High Stability Reference Option Signal Output Front or Rear Panel: The instrument related service note is a publication directed toward qualified service personnel and is available to all HP Service Centers and customers. Typical manufacturer of the part is a five-digit code. A21 Past To Present. Valid data for SM is: Signature Analysis Test 4.
3325A Synthesizer/Function Generator
Set A frequency to Hz, 1 kHz, and kHz and read ac voltage. Remove the filter and wash thoroughly with soapy water, rinse elcan, and air dry. Frequency Display and Resolution. Change A function to positive slope ramp and repeat Step f. Change oscilloscope horizontal control to. Change A amplitude to 40 V p-p. Set A frequency to 2 kHz and amplitude to 40 Vpp. The following accessory options are also available for the Model A: Observe the spectrum analyzer for spurious responses.
Each system can have only one system controller, even if several controllers have system control capability, A light in the center of the key indicates the present function. The HP-IB is a parallel bus of 16 active signal lines grouped into three sets according to function, to interconnect up to 15 instruments. The oscilloscope display should show a sweep that is essentially flat, dropping no more than 3.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
Ground the oscilloscope input and zero the trace on the center line. Enter 20 MHz in the A.
Amplitude is entered and displayed with 4-digit resolution. This manual also for: The preceding range of instruments contain the VCO circuitry shown in Figure Program examples using a specific Hewlett- Packard calculator as the system controller may be found in the Supplemental Programming Information, Appendix 3-A at the rear of this section.
An external reference may be used to phase Jock the internal 30 MHz reference see Paragraph Don’t show me this message again. This sync signal is always in phase with the output signal, with the sync transition occurring at the signal zero crossing, or when the signal crosses the dc offset voltage.
To rack mount with handles, the combination kit of item b Option must be used see Figure Frequency too large for function 4. Adjust Ref A3R30 for a counter display of The remaining program can then constitute a second string.
The following accessories are available for use with the Model A: Change A frequency to 1 kHz and kHz and repeat.