BMs – Top class digital multimeter made by Brymen. counts (5 4/5 digit), High Performance, User-oriented innovative features. Unbeatable 1µV. USER’S MANUAL BM BMCF 1 1) SAFETY This manual contains .. BRYMEN’s warranty does not apply to accessories, fuses, fusible resistors, spark . Thanks Lightages: + didn’t know TME also ships to the Netherlands prices are looking good indeed and no problem with customs since it’s.
|Published (Last):||7 April 2005|
|PDF File Size:||3.28 Mb|
|ePub File Size:||11.52 Mb|
|Price:||Free* [*Free Regsitration Required]|
Do not attempt a current measurement to any circuit where the open circuit voltage is above the protection fuse voltage rating.
BRYMEN TECHNOLOGY CORP, PICQ7.E – UL Product Spec Canada
Normal forward voltage drop forward biased for a good silicon diode is between 0. Press button momentarily bm859ct toggle between 5 full digits and 6 full digits Hz readings. However, if BRYMEN determines that the failure was caused by misused, altered, neglected, or damaged by accident or abnormal conditions of operation or handling, you will be billed for the repair. The new settings will be saved automatically to the non-volatile memory as power up default.
CMRR is the DMM’s ability to reject common mode voltage effect that can cause digit rattle or offset in voltage measurements. BRYMEN’s warranty does not apply to accessories, fuses, fusible resistors, spark gaps, batteries or any product which, in BRYMEN’s opinion, has been misused, altered, neglected, or damaged b,859cf accident or abnormal conditions of operation or handling.
Gm859cf the battery access door and thus the battery compartment up. Insert the banana plug K-type temperature bead probe Bkp60 standard accessory x 1 with correct polarities.
Brymen BM859CF Manuals
The meter beeps when new maximum or minimum reading is updated. You can also use a plug adapter Bkb32 Optional purchase with banana pins to K-type socket to adapt other standard K type mini plug temperature probes.
It can also be loaded way beyond seconds at 1A, and has a fast acting characteristic of below 0. If the meter is not to be used for periods of longer than 60 days, remove the battery and store it separately. Reverse the test leads connections reverse biased across the diode. Refer to the explanation in this Manual!
Line Level Frequency measuring function input sensitivity varies automatically with voltage or current function range selected. These voltage levels pose a potential shock hazard to the user. Note — Examples include household, office, b,859cf laboratory appliances. If the instrument is used in a manner not specified by the manufacturer, the protection provided by the instrument may be impaired. In measuring non-sinusoidal waveforms, however, significant errors can be introduced because of different scaling factors relating average to RMS values.
Only replace the blown fuse with the proper rating as specified in this manual.
This protection characteristic also matches our meter nicely. Battery replacement for models with battery access door: Do not measure any current that exceeds the current rating of the protection fuse.
If any defects are found, replace them immediately. The average value is then scaled upward calibrated to read the RMS value bgymen a sine wave. Distorted waveforms with the presence of DC components and harmonics may cause: Hold The hold function freezes the display for later view.
However, the continuity brymeen Jack Beep input warning features remain. Press the button momentarily again to step through the ranges. It’s a community-based project which helps to repair anything. Analog bar-graph The analog bar graph provides a visual indication of measurement like a traditional analog meter needle. This technique is to obtain the average value by rectifying and filtering the AC signal.
Brymen BMCF Manuals
Note — Examples include electricity meters and primary over-current protection equipment. The meter is intended only for indoor use.
If the reading shows zero, select higher sensitivity. Continuity function is convenient for checking wiring connections and operation of switches. A reading higher than that indicates a leaky diode defective.
A full wave rectified sine waveform is a good example, and the AC true RMS function will only give the AC component reading which is at The new setting will be saved automatically to the non-volatile memory as power up default. Relative Zero mode Relative Zero allows the user to offset the meter consecutive measurements with the displaying reading as the reference value.
Activities are specified as: The digital display shows OL if the diode is good. NMRR is typically specified in terms of dB decibel. Always set the instrument to the highest range and work downward for an unknown value when using manual ranging mode. Suspected open circuit voltage should be checked with voltage functions. It is bm89cf in detecting faulty contacts, identifying potentiometer clicks, and indicating signal spikes during adjustments.